Hot Weather Precautions for Laboratory RefrigeratorsJuly 19th, 2012
Unprecedented heat waves across much of the country place strains on laboratory refrigerators and laboratory freezers even if they are in air conditioned research labs and medical facilities. Power cutbacks and power failures due to excessive strain on the power grid and electrical storms should be anticipated by lab personnel. Provision should be made to protect costly or rare biological samples, vaccines, and other temperature-sensitive pharmaceuticals and products.
10 Tips on Proper Lab Fridge and Lab Freezer Operation
Your organization should have standard operating procedures for stocking and maintaining laboratory refrigerators and freezers. These are especially important when atmospheric conditions could lead to sub-par performance. Here are some useful tips to help ensure content protection in medical grade refrigerators and freezers during the dog days of summer. Or anytime for that matter. They are not listed in order of importance because all are important.
- Minimize access to the units. Every time a scientific refrigerator or scientific freezer door is opened ambient air intrudes and puts a strain on the compressor.
- Keep interiors filled. Full medical freezers and refrigerators operate more efficiently. Use freezer packs and water bottles to fill unused space.
- But not too full. Packed interiors inhibit circulation of refrigerated air.
- Storing content on door shelves and bins is not a good idea. Use the space for freezer packs or water bottles.
- Don’t crowd the units or stack paraphernalia on the top. Follow manufacturers’ instructions on positioning their units – such as a minimum 4 inches at the top, back and sides to allow ambient air to circulate.
- Clean condenser tubing and fins (and filters if the unit is so equipped) on a regular basis as dust inhibits efficient operation.
- Check door seals for leakage and take action if necessary.
- Test the high/low temperature alarming system on a regular basis.
- Log unit temperatures at the beginning and end of the day.
- Have a backup plan in case of a power outage. If you do not have an emergency generator make provisions for quick transfer of contents to operating units.
Laboratory refrigerators and freezers such as Nor-Lake and other brands available from Tovatech have several options in temperature control, display and alarming. But even the most sophisticated of these are not immune to malfunctions, as described in our blog post on how a major failure of backup systems caused severe loss at the Harvard Brain Tissue Resource Center.
Paying attention is the best insurance against loss. Contact the lab refrigerator and freezer experts at Tovatech for help in selecting and operating equipment best-suited to your needs.
Read more articles about : Lab Freezer, Lab Refrigerator.
Have Questions? Ask Us Now
|Previous Article||Determining Molecular Roles in Biogenesis||Next Article||Ultraviolet Exposure as a Countermeasure for Bioterror Threats|
No comments yet.